The HP A is capable of making highly accurate measurements. In order to .. you should use manual range to get to the lowest range possible for the. Hp A Pdf User Manuals. View online or download Hp A User Manual. View and Download HP A user manual online. A Multimeter pdf manual download.
|Published (Last):||22 April 2005|
|PDF File Size:||14.60 Mb|
|ePub File Size:||11.70 Mb|
|Price:||Free* [*Free Regsitration Required]|
Dc Input Resistance The Wait-for-trigger State Measurements will begin when the specified trigger conditions are satisfied after the INITiate command is received. There are two ways you can specify the null value.
Ieee Compliance Information Bits in the summary registers are not latched. To review the power-on display with all annunciators turned on, hold down as you turn on the multimeter. To Adjust The Carrying Handle This represents six full digits of resolution. You may want to disable the front-panel beeper mqnual certain applications.
| HP A repair and calibration worklog
Loading Errors ac Volts The rate is set RS to baud when the multimeter is shipped from the factory. When you have read all errors from the queue, the annunciator turns off.
Set the RS baud rate and parity see chapter 4. You can use the following commands to make dB measurements. This is particularly important if you are measuring resistances above k. Refer to the Service Guide for more information.
Chapter 3 Features and Functions Triggering Number of Samples Normally, the multimeter takes one reading or sample each time it receives a trigger from the selected trigger source if the multimeter is in the wait-for-trigger state. Front-panel Menu Reference Quick Start Quick Start One of the first things you will want to do with your multimeter is to become acquainted with its front panel. Error Messages Error Messages Errors are retrieved in first-in-first-out order.
Chapter 1 Quick Start To Select a Range To Select a Range You can let the multimeter automatically select the range using autoranging or you can select a fixed range using manual ranging. High-voltage Self-heating Errors Some of hobbyists were even upgrading reference to LTZ circuit for better A stability.
HP A power supplies. Note that the crest factor errors do not apply for input signals below For example, you can store such information as the last calibration date, the manuzl calibration due date, the instrument serial number, or even the name and phone number of the person to contact for a new calibration.
File:HP 34401A User Manual.pdf
Thermal Emf Errors Time worklog summary Image Ac Current Measurement Errors You may never have to change any of the measurement parameters discussed here, but they are provided to give you the flexibility you might need. To Prepare The Multimeter For Use The front-panel display will light up while the multimeter performs its power-on self-test. The multimeter uses the line to send a hold-off signal.
Chapter 3 Features and Functions Measurement Configuration 5 digits Dc Current Measurement Errors Page 77 remote interface bit definitions, SCPI HP-IB connector, 5 clearing, command summary, HP-IB selection, 92, status byte compliance information, language restrictions, 92, 94 bit definitions, data types, RS connector, 5, Programming Language Selection The questionable data enable register is cleared when: Throughout this manual, the following conventions are used for command syntax.
Low-level Measurement Errors Baud Rate Selection rs Number Of Triggers This may be useful in applications where you want to kanual the input to settle before taking a reading, or for pacing a burst of readings. The easiest way to program the multimeter for measurements is by using the MEASure? Errors are retrieved in first-in-first out order. Continuity H34401a Resistance Chapter 3 Features and Functions Measurement Configuration Continuity Threshold Resistance When measuring continuity, the multimeter emits a continuous tone if the measured resistance is less than the threshold resistance.